Powered USB Hub needed for Buxco Respiratory Hardware
When attempting to calibrate any of the Buxco Hardware sites (WBP, NAM, RC, and PFT) using FinePointe software, several error message may occur. The two most common error messages are:
- “Error: Instrument Not Found”
- “Error: Hardware is not functioning Properly”
Both of these error messages are caused by a flaw in the USB port in the Buxco hardware. The solution is to add a Powered USB Hub to the system.
We offer a Powered USB Hub under DSI Item# 271-0200-001.
This is Belkin High Speed USB Hub, item F5U237V1.
Can't find what you're looking for? Contact Us
Please sign in to leave a comment.